- Open Access
Towards developing a robot assistant for uterus positioning during hysterectomy: system design and experiments
© Yip et al.; licensee Springer. 2014
- Received: 2 July 2014
- Accepted: 8 September 2014
- Published: 24 December 2014
In traditional laparoscopic hysterectomy, an assistant is typically assigned to hold a uterus manipulator to facilitate the surgical procedures. The responsibility of the assistant is to position the uterus according to the primary surgeon’s instructions. Throughout the surgery, which typically lasts more than 90 min, the assistant has to support the uterus manipulator with his/her hands, which easily causes fatigue to the assistant and eventually affects the manipulation performance and threatens the patient. Moreover, the manipulation done by the assistant may not always be satisfactory from the primary surgeon’s point of view. Thus, having a robot assistant which can release the human assistant’s hands from the manipulation task and allow the primary surgeon to directly position the patient’s uterus can be a solution.
A four degrees of freedom (DOF) robot assistant is designed for positioning the patient’s uterus during hysterectomy. The robot assistant is composed of three parts, a 3-DOF robotic positioning arm, a 1-DOF motorized uterus manipulator, and a supporting stand of the robot. To improve safety, the remote center of motion (RCM) mechanism is applied to the design of the robotic positioning arm. The positioning arm generates a partial spherical workspace and allows the manipulation of uterus to be done in a decoupled manner. To enhance the ability of the robot assistant to perform anteversion and retroversion motions, the motorized uterus manipulator is designed. The robot assistant is mounted to the operating table through the supporting stand.
A prototype of the robot assistant is built, and experiments are conducted to verify the performance of the prototype. It is shown that the prototype is able to respond to the user’s instructions accordingly and manipulates the uterus of a manikin to a desired position. Performances of robot manipulation and human manipulation are also compared. It is shown that the robot assistant can retain the manikin’s uterus in a specified position in a more stable manner.
Our experimental study shows that the use of a robot is a feasible alternative to position the uterus. By using this type of system, costs can be reduced and safety can be improved.
- Robot-assisted surgery
- Laparoscopic hysterectomy
- Robot design
- Remote-center-of-motion mechanism
- Decoupled motion
- Prototype development
- Surgical tool manipulator
Excluding the pregnancy-related procedures, hysterectomy is the most commonly performed gynecological procedure . Since the 1990s, there has been an increasing trend in the number of hysterectomy procedures performed in laparoscopic approach . In the 2000s, with the introduction of the daVinci surgical system, surgeons started performing robot-assisted hysterectomy ,,. During a robot-assisted laparoscopic hysterectomy, the primary surgeon controls the robotic arms located at the patient’s bedside from the surgeon’s console, which is located away from the patient’s bed; the assistant surgeon at the patient’s bedside assists the primary surgeon in operating the uterus manipulator and using the accessory port of the daVinci system .
In , it is reported that the typical operative time for this kind of robot-assisted laparoscopic hysterectomy lasts no less than 190 min; in , it is reported that out of 16 cases, the median of the operative time is 242 min. However, for traditional laparoscopic hysterectomy conducted in the 2000s, it is reported that the median operative time is in general not more than 135 min . By comparison, it is observed that the operative time required to perform traditional laparoscopic hysterectomy (i.e., performed by the primary surgeon directly at the patient’s bedside) is in general shorter than those performed robotically with the daVinci system.
In terms of effectiveness, it is worth keeping the primary surgeon at the operating table to perform surgery. Besides, during this kind of robot-assisted surgery, operating the uterus manipulator is still the responsibility of the assistant surgeon. No matter in traditional or robot-assisted laparoscopic hysterectomy, asking an assistant surgeon to operate the uterus manipulator causes the problem that the manipulated position may not satisfy the primary surgeon’s need. Furthermore, if no supporting device is used to assist the assistant surgeon in holding the uterus manipulator, the assistant surgeon would suffer from the problem of fatigue as he/she is required to hold the uterus manipulator throughout the surgery. Fatigue may decrease the performance of the assistant surgeon and may eventually threaten the patient. Thus, a robotic system which is capable of assisting in this tool manipulation while keeping the primary surgeon at the operating table side during hysterectomy is necessary; this is precisely our aim in this article.
Existing surgical robots
Robots have been introduced to the operating theater in recent decades; the daVinci surgical system is a well-known example. The daVinci system is a multipurpose surgical robotic system designed for laparoscopic surgeries. Four robotic arms are included in the system for manipulating a laparoscope and other surgical tools during surgery. Though tele-surgery is allowed with the daVinci system, the primary surgeon who operates the robot is away distant from the patient’s bedside  due to its large footprint. Furthermore, as the robotic arms are designed for laparoscopic surgeries, the robotic arms are generally setup to operate above the patient’s abdomen; thus, when performing laparoscopic hysterectomy, an assistant is still required to position the patient’s uterus from the end of the operating table.
In general, surgical robots for minimally invasive surgery are mainly designed for endoscope manipulation or surgical tool manipulation during laparoscopic surgery in which several small incisions are opened on the patient’s abdomen and the endoscope and the surgical tools are inserted into the patient’s body through these incisions. Examples of robotic endoscope manipulators include AESOP , LARS , EndoAssist , and FreeHand .
EndoControl, the manufacturer of ViKY , a robot designed for endoscope or surgical tools manipulation, developed a system named ViKY UP  which aims at hysterectomy procedures. ViKY UP is a motorized uterus positioner which helps to manipulate the uterus manipulator during surgery. Though ViKY UP serves the aim of taking up the task of the assistant who is responsible for manipulating the uterus manipulator, the mechanism of the robot generates a cone-shaped workspace . With this configuration, joint motions are coupled when the surgeon would like to manipulate the patient’s uterus laterally or vertically.
In this article, we present a robotic prototype for uterus positioning during laparoscopic hysterectomy. Different from the majority surgical robots, the remote center of motion of our prototype is designed to be placed at the cervix (i.e., inside the patient’s body), which is more preferable in hysterectomy. With the use of the robot, safety can be improved as the robot can hold the uterus manipulator more stable than a human in general. Furthermore, a valuable manpower can be released with the use of the robot; thus, surgical costs can be reduced.
The robot should be able to grasp and hold the uterus manipulator securely.
The robot should have the ability to manipulate the uterus manipulator in a natural manner as an assistant surgeon does.
The robot should be able to receive commands from the primary surgeon and response correspondingly.
Anteversion/retroversion of uterus,
Laterally manipulation of uterus, and
Tensioning of uterus.
Design of the robot
The robotic system presented in this article is comprised of three parts, a robotic positioning arm, a motorized uterus manipulator, and a supporting stand of the robot.
Schematic of the robotic uterus manipulator positioning arm is shown in Figure 1. In the schematic diagram, the motorized uterus manipulator is used as the surgical tool manipulated by the positioning arm. The positioning arm has three degrees of freedom, yaw, pitch, and insertion. As in medical robotics, safety is the most important issue, a remote center of motion (RCM) mechanism  is used in the robotic positioning arm to improve safety. With the RCM mechanism, a point which is to be placed at or close to the patient’s cervix will be mechanically forced to remain stationary no matter how the arm moves. To create the remote center of motion, the three actuated joints are designed to satisfy specific geometric arrangement. Here, denote the joint for yaw motion by joint 1, the joint for pitch motion by joint 2, and the joint for insertion by joint 3. The geometric arrangement of the joints is as follow:
Joint 2 is a sliding joint which moves along a circular trajectory. The axis of rotation of joint 1 intersects with the center of the circular trajectory of joint 2. This intersection point gives the remote center of motion.
Joint 3 is a translational joint. The axis of translation of joint 3 intersects with the axis of rotation of joint 1 and the center of the circular trajectory of joint 2. The surgical tool considered as a straight rod lies on the axis of translation of joint 3 and intersects with the RCM.
The motorized uterus manipulator is a device which enhances the ability of the robot assistant in performing anteversion and retroversion motions. It is a device with a tilting tip which has a moving direction similar to joint 2. Here, we denote this joint by joint 4.
Actuation and mechanisms
To realize the motion of joint 2, pinion and rack mechanism is used. A block with a spur gear installed to the actuator is allowed to run along an arc-shaped guide rail with gear teeth on the rail as shown. A 20 W DC motor with gearbox is used as the actuator to ensure enough power can be delivered to drive the system. Using a motor with larger power than what it actually needs may also prevent the motor from damage due to overloading.
The structure of joint 2 is mounted on top of joint 1 directly, that is, joint 1 drives the entire structure including the arc-shaped guide rail of joint 2. Joint 1 is a revolute joint; the rotation motion from the actuator drives the structures mounted to it directly. A 20 W DC motor with gearbox is also used as the actuator for joint 1.
Joint 3 is mounted to the actuated block of joint 2. To realize the inserting motion, ball-screw mechanism is used. A 2 W DC motor with gearbox is used as the actuator as the ball-screw mechanism provides large reduction ratio, and the power required to drive the system is greatly reduced. An adaptor for grasping the surgical tool is attached to this joint. The adaptor is designed to hold uterus manipulators with a diameter of 8 or 12 mm.
Control system architecture
To verify the performance of the prototype, four experiments are conducted. Two experiments are conducted to verify the closed-loop response of the robot’s motion, and two are conducted to compare the manipulation performance of the robotic assistant with of a human assistant.
Velocity tracking experiment (auto-feed)
where v i represents the joint velocity and k i is a positive user-definable velocity gain.
Velocity tracking experiment (manual input from joystick)
As the robot is designed to be controlled and operated by the primary surgeon, this experiment is conducted in a manual-control mode. The objective of this experiment is similar to the auto-feed one, i.e., to verify the performance of the robot in responding to the user’s input commands.
where J i represents a discrete (i.e. −1,1,0) input provided by the joystick interface. The user is asked to control the robot by pressing the buttons on the control joystick which correspond to the robot joints and their moving directions.
Manipulation experiment (one-dimensional distance)
In this experiment, a medical manikin is used as the simulated environment. To emulate the laparoscopic camera, a webcam is used to obtain the visual feedback from the manikin. With the webcam, images with resolution of 640 × 480 pixel are obtained. An artificial marker is attached to the uterus as a visual feedback tracking point. In this experiment, a standard non-motorized uterus manipulator (produced by Apple Med ) is used as the surgical tool to position the uterus of the manikin.
On the robot side, the uterus manipulator is grasped by the robotic positioning arm. Human participants are asked to move the marker to a desired image configuration with the assistance of the robot and retain that position. The robot is controlled by the participants using a joystick.
To compare the performance of the robot with the performance of a human, the same experiment is conducted without the assistance of the robot. The human participants are asked to hold the uterus manipulator by themselves and keep a specified image position without the use of any supporting device. All participants involved in this experiment are without medical background. Short training for both manipulating with and without the robot is given to the participants before the experiment started.
where e is the pixel error, p d is the pixel position of the desired destination, and p m is the pixel position of the marker. In this one-dimensional experiment, p d and p m can be understood as the x-coordinates of the desired destination and the marker on the image plane, respectively.
In Figure 12, the plot on the left shows the measured pixel error obtained the during manipulation with the assistance of the robot; the plot on the right shows the measured pixel error obtained without the use of the robot assistant. From these results, it is shown that with the use of the robot, the participants are able to manipulate the uterus to a specified image position in a more stable way compared to the traditional hand manipulation.
Manipulation experiment (two-dimensional point control)
To further verify the performance of the robot, the manipulation experiment is extended to the two-dimensional image space. In this experiment, the desired position is specified by an image point instead of a line, as shown in Figure 11. The pixel errors in both x and y directions on the image plane are measured to evaluate the performances.
In this article, a four degrees of freedom robot assistant for uterus positioning during laparoscopic hysterectomy is designed and developed. This system is composed of a 3-degrees of freedom (DOF) uterus manipulator positioning arm, a 1-DOF motorized uterus manipulator, and a supporting stand which mounts the robot to the operating table. The 3-DOF uterus manipulator positioning arm adopts the RCM mechanism, which is designed to improve safety. Instead of placing the RCM out of the patient’s body as most surgical robots do, the RCM of the presented positioning arm is designed to be placed inside the patient’s body. Note that when applying to hysterectomy procedures, the RCM must be placed at or close to the cervix. The positioning arm has a spherical workspace, and the three actuated joints are designed to satisfy specific geometric arrangement so that the manipulation of the uterus can be achieved in a decoupled manner. A 1-DOF motorized uterus manipulator with a tilting tip is also designed and developed to enhance the ability of the robot assistant to perform anteversion and retroversion motions.
Experiments are conducted to verify the performance of the robot assistant, and the experimental results are presented. It is shown that the developed robot assistant can respond to the user’s commands and assist the user in uterus manipulation. Furthermore, when compare with solely human manipulation, the robot assistant shows the advantage of being able to retain a desired position in a more stable manner, which is precisely the goal of this prototype. Our robotic system can release a human assistant from the task of uterus positioning. This helps saving the valuable human resources in the operating theater. Besides, more reliable and stable positioning of uterus can also be achieved.
However, there are several disadvantages in our system, for example, space under the operating table is required when installing the robot assistant to the operating table. When encountering with operating tables with this space unavailable, the RCM of the robotic positioning arm will be unable to align with the patient’s cervix, which is undesirable. Also, due to the mechanical structure, the motion of joint 2 is comparatively not smoothly enough (see Figures 7 and 8) and the vibration frequency of the robot is high than that of human (see Figures 12, 13, and 14).
In the future, we would like to eliminate the necessity of under-bed space for installation so that the installation of robot can be more feasible when encountering with different operating tables. Also, we would like to improve the structure of the robotic system to increase its rigidity and reduce the system’s vibration. Improvements on the controller are also expected for the same purpose. One step further, image-based automatic control can also be implemented to provide more autonomy to the system.
Last but not the least, in our experiments, a manikin is used. Though medical manikin is the closest simulated environment when comparing to human, differences exist. Indeed, differences also exist among different patients. These differences may eventually affect the workspace and payload required for the robot. Thus, to obtain more realistic and reliable data for designing future robots, ex vivo experiments with cadavers and clinical trials are expected.
- Reynolds RK, Advincula AP: Robot-assisted laparoscopic hysterectomy: technique and initial experience. Am J Surg 2006, 191: 555–560. 10.1016/j.amjsurg.2006.01.011View ArticleGoogle Scholar
- Farquhar CM, Steiner CA: Hysterectomy rates in the United States 1990–1997. Obstet Gynecol 2002, 99: 229–234. 10.1016/S0029-7844(01)01723-9Google Scholar
- Beste TM, Nelson KH, Daucher JA: Total laparoscopic hysterectomy utilizing a robotic surgical system. J Soc Laparoendosc Surg 2005, 9: 13–15.Google Scholar
- Diaz-Arrastia C, Jurnalov C, Gomez G, Townsend CJr: Laparoscopic hysterectomy using a computer-enhanced surgical robot. Surg Endoscopy 2002, 16: 1271–1273. 10.1007/s00464-002-8523-5View ArticleGoogle Scholar
- Hysterectomy: procedures, complications, and alternatives. Nova Science Publishers, Inc., New York,; 2011.Google Scholar
- Guthart GS, Salisbury, J Jr (2000) The intuitive telesurgery system: overview and application In: Robotics and automation, 2000. Proceedings. ICRA ‘00. IEEE international conference on, vol. 1, 618–621.Google Scholar
- Sackier JM, Wang Y: Robotically assisted laparoscopic surgery. Surg Endoscopy 1994, 8: 63–66. 10.1007/BF02909496View ArticleGoogle Scholar
- Taylor RH, Funda J, Eldridge B, Gomory S, Gruben K, LaRose D, Talamini M, Kavoussi L, Anderson J: A telerobotic assistant for laparoscopic surgery. IEEE Eng Med Biol Mag 1995, 14: 279–288. 10.1109/51.391776View ArticleGoogle Scholar
- Aiono S, Gilbert JM, Soin B, Finlay PA, Gordan A: Controlled trial of the introduction of a robotic camera assistant (endoassist) for laparoscopic cholecystectomy. Surg Endoscopy 2002, 16: 1267–1270. 10.1007/s00464-001-9174-7View ArticleGoogle Scholar
- Sharma D, Brown C, Kouriefs C, Sood H, Grange P, Patel H: Initial experience with the freehand robotic camera holder in laparoscopic urology. J Endourol 2009, 23(1):A249.Google Scholar
- Berkelman P, Ma J: A compact modular teleoperated robotic system for laparoscopic surgery. Int J Robot Res 2009, 28: 1198–1215. 10.1177/0278364909104276View ArticleGoogle Scholar
- Swan K, Kim J, Advincula AP: Advanced uterine manipulation technologies. Surg Tech Int 2010, 20: 215–220.Google Scholar
- Long JA, Cinquin P, Troccaz J, Voros S, Berkelman P, Descotes JL, Letoublon C, Rambeaud JJ: Development of miniaturized light endoscope-holder robot for laparoscopic surgery. J Endourol 2007, 21: 911–914. 10.1089/end.2006.0328View ArticleGoogle Scholar
- Kuo C-H, Dai J: Robotics for minimally invasive surgery: a historical review from the perspective of kinematics. In International symposium on history of machines and mechanisms. Edited by: Yan H-S, Ceccarelli M. Springer, Netherlands; 2009:337–354. 10.1007/978-1-4020-9485-9_24View ArticleGoogle Scholar
- Ng C, Chern B: Total laparoscopic hysterectomy: a 5-year experience. Arch Gynecol Obstet 2007, 276: 613–618. 10.1007/s00404-007-0385-6View ArticleGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/4.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly credited.