A two-axis cable-driven ankle-foot mechanism
© Ficanha et al.; licensee Springer. 2014
Received: 26 August 2014
Accepted: 15 October 2014
Published: 2 November 2014
This paper describes a novel cable-driven ankle-foot mechanism with two controllable degrees of freedom (DOF) in dorsiflexion-plantarflexion (DP) and inversion-eversion (IE). The presented mechanism is a proof of concept to demonstrate feasibility. Ankle kinematic measurements demonstrate that ankle IE rotations during a step turn are significantly different from walking on a straight path. This suggests that the ankle-foot mechanisms used in prostheses, exoskeletons, and bipedal robots can be improved by controlling a second degree of freedom in the frontal plane. The proposed prototype mechanism is described in detail, and its design considerations and parameters are presented. The mechanism is capable of producing trajectories similar to the human ankle during a step turn. The device shows passive mechanical impedance close to the human ankle mechanical impedance, allowing its mechanical impedance to be controlled using an impedance controller. The presented mechanism is capable of providing key mechanical characteristics similar to the human ankle, including power, range of motion, and weight, suggesting the feasibility of this design concept.
KeywordsAnkle mechanism Ankle impedance Two-DOF ankle-foot Turning Ankle rotations Cable-driven prosthesis Human ankle kinematics Multi-axis ankle-foot prosthesis Powered lower extremity prosthesis
Cable-driven mechanisms have been used on upper extremity exoskeletons and assistive robots such as prosthetic hands. In such mechanisms, pulling the cables results in motion, the same way that the muscles pull tendons in the musculoskeletal system. Exploiting a cable-driven system allows the DC motors to be placed at locations that are not necessarily close to the mechanisms’ joints, hence, assisting in maintaining a low profile and reducing the size and weight of the device. One example of such design is lower extremity powered exoskeleton (LOPES) with eight actuated degrees of freedom at the hip, knee, and pelvis .
A similar cable-driven design concept can be followed in the lower extremity prostheses design to extend the controllable degrees of freedom (DOFs) beyond state of the art. Currently available ankle-foot prostheses are capable of controlling the ankle joint in only a single DOF in the sagittal plane. These prostheses improve the quality of life of amputees and are designed to increase mobility. However, turning plays a major role in activities of daily living and account for an average of 25% of steps, ranging from 8% to 50% of all steps depending on the activity . Turning requires modulation of ankle impedance in the sagittal and frontal planes to control lateral and propulsive ground reaction forces in order to accelerate the body center of mass along the gait path, resulting in increased lateral and propulsive impulses when compared to a straight step . As a result, turning strategies for amputees and non-amputees are different. Non-amputees rely on hip movement in the frontal plane and moment generation at the ankle joint, while amputees using passive prostheses rely on hip extension in the sagittal plane to compensate for the lack of forward propulsion from their passive prosthesis -. This data suggests that the next generation of ankle-foot prosthesis can benefit from designs that are capable of functioning in both the frontal and sagittal planes. A cable-driven mechanism seems to be a suitable solution to overcome the weight, size, and profile constraints while introducing a second actuation component. An ankle-foot mechanism capable of impedance modulation and power generation in both the sagittal and fontal planes allows the user to turn in arbitrary directions and walk on slopes that will increase their agility.
When physical systems interact with each other, they behave either as an impedance (i.e., accepts external motion inputs and generates force outputs) or an admittance (i.e., accepts external force inputs and generates motion outputs) . The human ankle functions by adjusting its mechanical impedance and capability of producing net positive work during push-off. These capabilities have inspired the design of several new ankle-foot prostheses. One example is a transfemoral prosthesis with powered ankle and knee joints in the sagittal plane developed by Sup et al. where the controller is capable of regulating the impedance of each joint as needed -. SPARKy 1 and 2 have been developed by Hitt et al. as a tendon-actuated ankle-foot prostheses capable of producing the necessary push-off moments during walking and running, respectively ,. Additionally, Au et al. developed the BiOM, an ankle-foot prosthesis with a finite state machine to determine the gait phase and generate the appropriate kinetic output -. All three of the aforementioned devices are capable of increasing mobility in the sagittal plane by producing net positive work in the sagittal plane. Finally, Bellman et al. conceived SPARKy 3, an ankle robot with two DOFs; however, a prototype design has not been developed . This leads to the fact that a design to address the weight and size limitations is necessary to develop the next generation of ankle-foot prostheses capable of properly functioning in both the sagittal and frontal planes. These additional capabilities will allow a more natural gait, especially during turning and traversing inclined planes.
The paper is structured as follows: first, the mechanism design is described in details. Next, a trajectory control and instrumentation necessary for a torque feedback is described. Finally, the evaluation experiments are described showing the mechanism capability for reproducing the human ankle kinematics during a step turn. Additionally, a comparison of the passive mechanical impedance of the mechanism’s ankle with the human ankle is provided.
Ankle kinematics and design considerations
Ankle kinematics in the frontal plane for the right ankle during the straight steps (Figure 2A) in both slow and fast speeds did not differ; however, during a step turn (Figure 2B), there was a significant increase in the ankle inversion. At straight step and low speed, the ankle was in 2.8° eversion during terminal stance, while during the step turn, it changed to 10.5° inversion. By increasing the speed, 2.5° eversion and 14.3° inversion were observed for straight step and step turn, respectively. The changes were equivalent to 248% and 303% of the straight step average range of motion (ROM) in the frontal plane for the slow and fast speed tests (5.37° ±0.6° and 5.52° ±0.8°, respectively). Additionally, the average maximum angular velocities among the participants during the fast speed tests were 120°/s in the sagittal plane and 101°/s in the frontal plane. The maximum angular displacements were 22° in plantarflexion and 20° in inversion. These values are important since an anthropomorphic ankle-foot mechanism should be able to reproduce them. There was also an increased rotation of the ankle in IE during turning. This demonstrated the significance of controllable frontal plane rotations in an ankle-foot mechanism for improved maneuverability.
In the transverse plane, the changes in ankle rotations were not as significant as in the frontal plane. The average ankle rotations during the sidestep cut (Figure 2A) decreased during turn at the terminal stance by −32% with respect to the average ROM during the straight walk in the transverse plane. However, hip rotations in the transverse plane also need to be accounted in the relative motion of the body with respect to the foot. Further investigation is required to quantify the contributions of the hip and ankle to the transverse plane moments and angles during locomotion before implementing it in the design of ankle-foot mechanisms.
The prototype design consists of two DC motors and planetary gear heads (A) powered by two motor controllers (B) connected to two quadrature encoders (C) (Figure 1). Two cable drums (D) transfer the required torque to the ankle through the shock-absorbing nylon rope (E). The rope is securely attached to the cable drums to avoid slippage. A universal joint (F) connects the pylon to the foot and is supported by an elastomer to provide passive stiffness and damping to the ankle. The cable is directed to the foot with pulleys (G). The cable is attached to a carbon fiber plate (H), which is connected to a commercially available prosthetic foot (Otto Bock Axtion®) (I). In the rear side of the carbon fiber plate, the cable is mounted to both sides of the longitudinal axis of the foot. At the front side of the carbon fiber plate, the cable passes through a pulley (J). Torque feedback, which will be explored later, is provided by six strain gauges in the foot (K) using two strain gauge amplifiers (L). An analog to digital converter (M) is connected to a remote computer and is used to acquire the sensors data and provide the motor controllers’ inputs.
The proposed design benefits from using two motors simultaneously. The motor torques can be combined to produce DP torques and subtracted to produce IE torques. Considering the fact that more power is required in the sagittal plane than the frontal plane during walking, this approach is more appropriate than employing a larger motor for DP and one smaller motor for IE. Additionally, it does not require any extra hardware other than the universal joint and the pulley at the ankle.
The cable-driven design, besides the ability to control the ankle in two DOFs, can provide significant flexibility for managing the ankle-foot inertia. The motors and gear boxes can be located remotely and still provide the necessary characteristics and user preferences. The actuation system has the potential to move up or down on the pylon to accommodate the users’ residual limb or to optimize the weight distribution and inertia characteristics.
Selection and design of the active components
For an average able-bodied human weighing 80 kg, the energy consumption required at each step is 36 J (250 watts peak power)  and torque as high as 140 Nm  during walking. These amounts are 35% higher for an individual with transtibial prosthesis ,, and the mechanism is estimated to have 40% losses, resulting in an anticipated peak power consumption of 470 watts, an energy consumption of 68 J, and a peak torque of 264 Nm in the sagittal plane. Also, the device needs to generate 120 deg/s for the sagittal plane and 101°/s for IE, as estimated from the gait experiment described previously.
where τDP and τIE are torques in DP and the IE, respectively, and Fpull − left and Fpull − right are the pulling forces from the left and right motors.
Due to the design geometry, when the foot deviates from the neutral position, the cables do not connect to the carbon fiber plate at a 90° angle, resulting in losses of the generated torque. These losses are taken into the account by cos(θ DP ) and cos(θ IE ) terms in Equations 3 and 4, respectively. At the maximum angular displacements observed in the gait experiment, the prototype shows a 7% loss in DP torque at −21.6° plantarflexion (maximum DP angle) and a 6% loss in IE at 19.7° inversion (maximum IE angle). These nonlinear losses are small but should be considered during the control of the mechanism.
where and are the cable velocities pulling from the left and right motors, respectively. The motors (at 0.275 Nm torque and 10,000 RPM) are capable of generating 256 Nm at 128°/s in DP and 64 Nm at 256°/s in IE. These values are close to the estimated 264 Nm of required torque in the sagittal plane based on the maximum torque required for walking  and the expected losses in the prosthesis and close to the angular velocities of 120°/s for DP and 101°/s for IE measured in the gait experiment described previously.
Selection and design of the passive component
The shock-absorbing nylon rope stretches to provide shock absorption while being soft to the touch. Nylon ropes are strong and flexible which makes it ideal for the design of this prototype. The carbon fiber plate is 3.175 mm thick and is a fundamental component of the design. It has been reported that the average plantarflexion stiffness at 50% MVC is 143 Nm/° and is near constant from 30% to 80% MVC ; thus, the carbon fiber plate was designed to provide a stiffness comparable to these amounts in the sagittal plane. The carbon fiber plate acts as a spring connected in series between the cable and foot. The cable needs to be always in tension to assure the proper control over the foot, causing the carbon fiber plate to be always under a bending moment.
Maximum force, torque, and range of motion
A Kistler® Type 5233A force plate was used to measure the maximum amount of force produced by the ankle-foot mechanism with a step input in plantarflexion. The mechanism’s pylon was constrained in the z direction with the intention to measure the torque at the ankle while DC motors were applying the stall torques. The device was capable of generating a lift force of 712 N, which corresponds to 103.2 Nm torque. The maxim IE torque can be calculated directly from the DP torque and the mechanism’s geometry to an estimated 25.8 Nm.
In the current configuration, the maximum range of rotation recorded from the position encoders was 49° in DP and 98° in IE, due to the range of rotation of the cable drums. Both DP and IE ranges are within the set goals. Testing the ROM in IE, it was found that the IE motion becomes unstable at angles above 62° (if forced by an external torque to this angle) similar to rolling of the ankle, which is an occurring injury among sport players and is in inversion . In the prototype mechanism, ankle roll happens when either points A or B (Figure 3) crosses the Y-axis. At this configuration, the tension in the cables, which are applying a torque against the disturbance force, change the direction of the torque and makes the ankle unstable. The maximum expected rotation observed in IE during the gait tests using the camera system was 19.7° and thus instability should not be a problem during reproduction of a straight walk or turn.
Instrumentation and control
Ankle torque and angle feedback
To actively regulate the ankle impedance, torque and angle feedbacks are required. The prototype is equipped with strain gauges mounted to the foot, which can be correlated to the external torques and will be used in our future work for impedance modulation of the prototype. Strain gauges are commonly used in load cells to measure the strains in structures due to external forces. The strain gauges change the resistance as they stretch or contract, and their change in resistance is correlated to the strain of the object they are attached to and consequentially the force or torque applied to the object. Strain gauges are typically wired in a Wheatstone bridge configuration using four strain gauges (some of the strain gauges can be replaced by resistors). The increase or decrease in resistance of two of the strain gauges (placed in opposite sides of the bridge) causes the output voltage of the bridge to increase or decrease, respectively. The opposite is also true for the other two strain gauges where the increase or decrease in resistance decreases or increases the output voltage of the bridge, respectively.
For estimating the torque in IE, two strain gauges were attached to the top surface of the foot as shown in Figure 4B. The other two strain gauges were attached to an inert piece of carbon fiber but could also be replaced by two resistors with resistance identical to the strain gauges. The strain gauges were placed on the outside edges of the foot and were on the same side of the Wheatstone bridge (Figure 5B); hence, the difference in strains caused an increase or decrease in voltage at the bridge. The voltage could be correlated to the IE torque in the foot when the front of the foot was contacting the ground (e.g., push-off). This configuration made the bridge insensitive to the torque in the sagittal plane, since the strain gauges were in the opposite ends of the bridge. Therefore, if they both contracted or stretched by the same amount as it would happen in the presence of a DP torque, the output would not be affected. This feature was important since it was necessary to have the frontal plane torque estimation to be decoupled from the sagittal plane torque.
To correlate the strain gauge readings to the actual disturbance torques, a Kistler® Type 5233A force plate was used to measure the external force applied during quasi-static loading tests. The tests consisted of loading the foot in different configurations and recording the applied force and the corresponding strain measurement. The tests were performed in 1 - plantarflexion by applying a load when the heel was in contact with the ground, 2 - dorsiflexion by applying a load when the forefoot was in contact with the ground, 3 - eversion by applying a load when the right edge of the forefoot was in contact with the ground, and 4 - inversion by applying a load when the left edge of the forefoot was in contact with the ground. From the external forces, the geometry of the foot, and the strain measurements, the applied torques were calculated. It is important to note that in the sagittal plane, the proportional factor between the external force and the strains measured at heel loading and forefoot loading were not the same, since the strain gauges were attached to two different areas of the foot. The proportional factors for the strain gauges at heel loading and forefoot loading were estimated at 1.41 and 19.52 Nm/volt, respectively. As a result, different proportional factors needed to be used depending if the strain measurement was positive or negative. In the frontal plane, the proportional factors for inversion and eversion torques were closer (4.43 and 3.55 Nm/volt, respectively), which was expected since the foot is near symmetrical about its sagittal plane.
The overall controller consists of two identical PD controllers to estimate the appropriate motor voltages. The input to the left motor controller is the sum of DP and IE angles, while the input to the right motor controller is the difference between the DP and IE angles. The desired angular displacements were the ankle angles measured during the gait experiment using the camera system that were stored in a lookup table to be reproduced by the ankle-foot mechanism for evaluation purposes as discussed in the following sections.
Evaluation of the ankle-foot mechanism
The aim of this evaluation study is to show that the mechanism is capable of reproducing the human ankle rotations during straight walk and turning. The input of the controller is the time history of a representative subject’s ankle angles in DP and IE recorded with a camera system and stored in a lookup table. The output is the ankle angles calculated using the quadrature encoders.
Due to the physical characteristics of the mechanism, small angular differences between the left and right motors caused larger changes in the foot rotations, making the system more sensitive to disturbances and noise in the frontal plane compared to the sagittal plane. Note that the current prototype was developed as a proof of concept to validate the design kinematics and uses low-cost brushed DC motors and gear boxes with limited bandwidth, which are the cause of the observed time delay. Therefore, faster motors and sensors with lower noise levels will be used in future designs. Nevertheless, both plots indicate that the mechanism is capable of reproducing the ankle rotations of a human subject during step turn with desirable accuracy.
Mechanical impedance of a system is the relationship between the displacement input and its force outputs. Control strategies have been developed for impedance control of active prostheses ,, where higher level control identifies the gait cycle and the lower level control regulates the actuators impedance characteristics. Ankle-foot robots can be designed to show passive and active mechanical impedance similar to the human ankle. In a stationary position, the passive mechanical impedance is dependent only to the passive components of the device. If the passive mechanical impedance of the mechanism is similar to the human ankle mechanical impedance, it may allow the impedance controller to easily regulate the active mechanical impedance of the mechanism to the target impedance for each specific gait cycle.
Sixty seconds of data were sampled at 200 Hz yielding 12,000 samples for both Anklebot and for the combined Anklebot and ankle-foot mechanism. Welch’s averaged, modified periodogram method of spectral estimation, as implemented in MATLAB, was used to estimate one-sided auto and cross-power spectral densities of the torque and angle sequences in the DP and IE directions. A periodic Hamming window with a length of 512 samples was used with an FFT length of 1,024 samples, yielding a spectral resolution of 0.195 Hz. Power spectral density functions were estimated by averaging their values calculated from 45 data windows with 50% overlap (256 samples).
Results and discussion
Ankle rotations during straight walk and step turn were measured using a camera system to study the ankle function in the frontal plane during the turn. The ankle rotations in the frontal plane during straight walk in both slow and fast speeds were close; however, there was a significant increase in the ankle’s rotation in the IE direction during the step turn. The ankle angles in IE during the push-off increased from 2.8° eversion during straight walk at low speed to 10.5° inversion during the step turn at slow speed. These values were 2.5° eversion and 14.3° inversions during the step turn for straight walk and step turn, respectively. These changes were equivalent to 248% and 312% of the straight step average ROM in the frontal plane for the slow and fast speed tests, respectively. The changes observed in the angular displacement of the ankle in the frontal plane during step turn, when compared to straight walk, showed the importance of the IE rotation during turning.
Mechanical characteristics of the ankle mechanism
The prototype ankle-foot mechanism was designed to control the foot in two DOFs. The mechanism uses two identical motors to generate torques in the sagittal plane and in the frontal plane. When both motors rotate in the same direction, they generate DP motion at the foot. IE motion is also generated when the motors move in opposite directions. Sagittal plane torque requirements are larger than the frontal plane torque requirements, and the design benefits from its combined power from both motors to generate DP torques and difference in torques to generate IE torques, instead of using one large motor for DP and one smaller motor for IE. The cable design also allows for placement of the motors at different places on the pylon. This allows the design to be easily adjusted for amputees with different residual limb length or to optimize weight distribution. Additionally, the design may be significantly improved using Bowden cables, allowing the motors and gear boxes to be mechanically decoupled from the pylon and placed elsewhere. This will allow for great flexibility on the design and customization of the prostheses based on individual user needs, especially for amputees with long residual limbs. Also, having the flexibility of placing the motors and gearboxes, which carry the most of the weight of the mechanism, away from the distal end of tibia would allow for a lightweight prosthesis that in theory can result in more symmetric gait patterns.
The prototype ankle mechanism was designed to match the mechanical characteristics of the human ankle. This included power, range of motion, and weight. The maximum power consumption, based on the energy expenditure of humans and the expected losses in the mechanism, is 470 watts. The motors chosen for the ankle mechanism have a combined power of 480 watts. The motors are rated at 0.275 Nm torque at 10,000 RPM, which results in 256 Nm at 128°/s in DP and 64 Nm at 256°/s in IE. These values are similar to the estimated 264 Nm of required torque in plantarflexion, and the angular velocities are close to 120 and 101°/s required for the sagittal plane and the frontal plane motion, respectively. Using a force plate, the maximum lifting force of the ankle mechanism was found to be 712 N in the sagittal plane which is the equivalent of the weight of a 73-kg person.
Impedance controllers have been used in active prosthesis to control the position and torque generated by the device. An impedance controller requires both position feedback and force feedback of the interactive forces between the device and the environment. The ankle mechanism in its current form uses quadrature encoders at the cable drums for position feedback. This provides an estimate of the ankle angles but does not account for the nonlinearities of the device caused by the rotation of the ankle, or the deflection of the carbon fiber plate, and stretch of the nylon rope. This can be remedied by using the encoders to measure the rotation of the foot directly, instead of the cable drums’ rotations. The torques generated from the mechanism can be measured using strain gauges at the foot, as described in this paper. However, the nonlinearities caused by the rotation of the foot also affect the generated torque, which need to be accounted in the controller design in the future. This will allow the use of ground reaction torques feedback and the angular displacements of the foot in the implementation of impedance controllers for the two-axis ankle mechanism in both the sagittal plane and the frontal plane.
Using a simple position controller, the recorded ankle rotations of a subject were reproduced by the prototype ankle mechanism. It was observed that the device was capable of tracking the human ankle motion in both the sagittal plane and the frontal plane. Since the mechanism was designed to use the difference in motor torques and angles to generate frontal plane torques and angular displacements, respectively, the design was found to be more sensitive to disturbances and noise in IE in comparison to DP.
The impedance estimation test showed that the quasi-static stiffness (below 1 Hz) of the prototype mechanism was 39.5 dB (94 Nm/radian) in the sagittal plane with relatively linear impedance and phase characteristics up to 5 Hz, which is the frequency of interest. The linearity of the system, shown as the average coherence of the estimated impedance, was 0.92. These values were similar to the ankle stiffness of a human subject in the sagittal plane during the muscle co-contraction test and higher than the subjects’ quasi-static stiffness with relaxed muscles. In the frontal plane, the mechanism impedance was between the active and passive stiffness of the human subject’s ankle, with a stiffness of 24.24 dB (16 Nm/radian). Considering the variation in ankle stiffness among individuals, these experiments show that the mechanism can be designed with a passive impedance similar to the human ankle. This will facilitate controlling the impedance of the mechanism through an impedance controller to the necessary values based on the gait requirements.
The current prototype was developed as a proof of concept to validate the design kinematics. It uses low-cost brushed DC motors and compatible gear boxes, resulting in low bandwidth of the motors and gear box, limited efficiency and power to weight ratio of the mechanism. Even so, the mechanism was capable of reproducing the ankle rotations of a human subject during step turn, adequate torques, and desirable mechanical impedance similar to a human ankle, while weighing 3 kg. Therefore, using faster, brushless motors and sensors with lower noise levels in future designs will increase the efficiency, power, and bandwidth, while decreasing the device’s weight.
In this paper, we described a novel two-axis, cable-driven ankle-foot mechanism. First, we described an experiment using a motion capturing camera system to study the human ankle during straight walk and turning, which showed that the inversion and eversion of the human ankle are significant during turning. This suggested that the design of ankle-foot mechanisms could be improved by controlling the degree of freedom of ankle mechanisms in the frontal plane. The proposed mechanism was described in detail, and its design parameters and considerations were presented. Moreover, a method for estimating the external torques applied to the ankle using strain gauges was presented, which is necessary in the development of impedance controllers. A simple position controller was implemented and used to demonstrate the capability of the device in tracking the same motion as the human ankle during a turning maneuver. It was shown that the prototype was capable of generating enough torque to lift a 73 kg person. An experiment was performed to explore the mechanism’s passive mechanical impedance. The device showed passive mechanical impedance close to the impedance of a human subject, which prompts the feasibility of regulating the ankle-foot mechanism mechanical impedance using an impedance control. The presented mechanism was capable of providing key mechanical characteristics similar to the human ankle, including power, range of motion, and weight, while allowing instrumentation for the development of an impedance controller in both dorsiflexion-plantarflexion and inversion-eversion.
Degrees of freedom
Lower extremity powered exoskeleton
Maximum voluntary contraction
Range of motion
This material is based upon work supported by the National Science Foundation under CAREER Grant No. 1350154.
- Veneman JF, Kruidhof R, Hekman EEG, Ekkelenkamp R, Asseldonk EHFV, Hvd K: Design and evaluation of the LOPES exoskeleton robot for interactive gait rehabilitation. IEEE Trans Neural Syst Rehabil Eng 2007, 15(3):379–386. 10.1109/TNSRE.2007.903919View ArticleGoogle Scholar
- Glaister BC, Bernatz GC, Klute GK, Orendurff MS: Video task analysis of turning during activities of daily living. Gait Posture 2007, 25(2):289–294. 10.1016/j.gaitpost.2006.04.003View ArticleGoogle Scholar
- Glaister BC, Orendurff MS, Schoen JA, Bernatz GC, Klute GK: Ground reaction forces and impulses during a transient turning maneuver. J Biomechanics 2008, 41(4):3090–3093. 10.1016/j.jbiomech.2008.07.022View ArticleGoogle Scholar
- Ventura JD, Segal AD, Klute GK, Neptune RR: Compensatory mechanisms of transtibial amputees during circular turning. Gait Posture 2011, 34: 307–312. 10.1016/j.gaitpost.2011.05.014View ArticleGoogle Scholar
- Orendurff MS, Segal AD, Berge JS, Flick KC, Spanier D, Klute GK: The kinematics and kinetics of turning: limb asymmetries associated with walking a circular path. Gait Posture 2006, 23(1):106–111. 10.1016/j.gaitpost.2004.12.008View ArticleGoogle Scholar
- Hase K, Stein RB: Turning strategies during human walking. J Neurophysiol 1999, 81(6):2914–2922.Google Scholar
- Segal AD, Orendurff MS, Czerniecki JM, Schoen J, Klute GK: Comparison of transtibial amputee and non-amputee biomechanics during a common turning task. Gait Posture 2011, 33(1):41–47. 10.1016/j.gaitpost.2010.09.021View ArticleGoogle Scholar
- Hogan N: Impedance control: an approach to manipulation: part I—theory. J Dyn Sys Meas Control 1985, 107(1):1–7. 10.1115/1.3140702View ArticleGoogle Scholar
- Goldfarb M: Powered robotic legs – leaping toward the future. Bioengineering, National Institute of Biomediacl Imaging and; 2010.Google Scholar
- Sup F, Bohara A, Goldfarb M: Design and control of a powered transfemoral prosthesis. Int J Rob Res 2008, 27: 263–273. 10.1177/0278364907084588View ArticleGoogle Scholar
- Sup F, Varol HA, Mitchell J, Withrow TJ, Goldfarb M: Preliminary evaluations of a self-contained anthropomorphic transfemoral prosthesis. IEEE ASME Transaction of Mechatronics 2009, 14(6):667–676. 10.1109/TMECH.2009.2032688View ArticleGoogle Scholar
- Sup F: A powered self-contained knee and ankle prosthesis for near normal gait in transfemoral amputees. Vanderbilt University, Nashville, Tennessee; 2009.Google Scholar
- Hitt J, Merlo J, Johnston J, Holgate M, Boehler A, Hollander K, Sugar T (2010) Bionic running for unilateral transtibial military amputees. In: 27th Army Science Conference (ASC). Orlando, FloridaGoogle Scholar
- Hitt JK, Sugar TG, Holgate M, Bellman R: An Active foot-ankle prosthesis with biomechanical energy regeneration. J Med Devices 2010, 4(1):011003. 10.1115/1.4001139View ArticleGoogle Scholar
- Au SK: Powered ankle-foot prosthesis for the improvement of amputee walking economy. Massachusetts Institute of Technology, Cambridge, MA; 2007.Google Scholar
- Au S, Herr H: Powered ankle-foot prosthesis. Robot Automation Mag 2008, 15(3):52–59. 10.1109/MRA.2008.927697View ArticleGoogle Scholar
- Au S, Weber J, Herr H: Powered ankle-foot prosthesis improves walking metabolic economy. IEEE Trans Robot 2009, 25(1):51–66. 10.1109/TRO.2008.2008747View ArticleGoogle Scholar
- Au SK, Herr H, Weber J, Martinez-Villalpando EC: Powered ankle-foot prosthesis for the improvement of amputee ambulation. International Conference of the IEEE, Engineering in Medicine and Biology Society, Lyon, In; 2007.View ArticleGoogle Scholar
- Herr HM, Grabowski AM: Powered ankle-foot prosthesis improves metabolic demand of unilateral transtibial amputees during walking. Amer. Society of Biomechanics, Long Beach, CA, In; 2010.Google Scholar
- Bellman RD, Holgate MA, Sugar TG, Bellman RD, Holgate MA (2008) Sugar TG SPARKy 3: design of an active robotic ankle prosthesis with two actuated degrees of freedom using regenerative kinetics. In: Proceedings of the 2nd Biennial IEEE/RAS-EMBS International Conference on Biomedical Robotics and Biomechatronics. Scottsdale, AZ, USA, pp 511–516Google Scholar
- Samuel K, Au JW, Hugh H (2007) Biomechanical design of a powered ankle-foot prosthesis. In: Paper presented at the International Conference on Rehabilitation Robotics, Noordwijk. The NetherlandsGoogle Scholar
- Rao SS, Boyd LA, Mulroy SJ, Bontrager EL, Gronley JK, Perry J: Segment velocities in normal and transtibial amputees: prosthetic design implications. IEEE Trans Rehabil Eng 1998, 6(2):219–226. 10.1109/86.681188View ArticleGoogle Scholar
- Sinkjaer T, Toft E, Andreassen S, Hornemann BC: Muscle stiffness in human ankle dorsiflexors: intrinsic and reflex components. J Neurophysiol 1998, 60(3):1110–1121.Google Scholar
- Saripalli A, Wilson S: Dynamic ankle stability and ankle orientation. In Paper presented at the 7th Symp. Footwear Biomech. Conf, Cleveland, OH; 2005.Google Scholar
- Eilenberg MF, Geyer H, Herr H: Control of a powered ankle-foot prosthesis based on a neuromuscular model. IEEE Trans Neural Syst Rehabil Eng 2010, 18(2):164–173. 10.1109/TNSRE.2009.2039620View ArticleGoogle Scholar
- Rastgaar M, Ho P, Lee H, Krebs HI, Hogan N (2010) Stochastic estimation of the multi-variable mechanical impedance of the human ankle with active muscles. In: ASME Dynamic Systems and Control Conference. Boston, MAView ArticleGoogle Scholar
- Rastgaar M, Ho P, Lee H, Krebs HI, Hogan N (2009) Stochastic estimation of multi-variable human ankle mechanical impedance. In: ASME Dynamic Systems and Control Conference. Hollywood, CAView ArticleGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/4.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly credited.